High Frequency dental x ray machine MSLK04
New x-ray unit MSLK04 Product Parameters
a) The rated parameters of x-ray production device
◆Tube voltage: 70kv (fixed)
◆Tube current: 1MA (fixed)
◆High voltage generating circuit: High frequency switch
◆X-ray control method: Microprocessor control
◆Time setting range: 0.05-1.2 seconds
b) Weight: 1.95kg
c) Size: 215*140*160 mm
d) X-ray tubing
◆Type: bipolar fixed x-ray tube
◆X-ray tube Model: DG-073B-DC
◆The size of x-ray tube focal spot: 0.3*0.3mm
◆ANODE heating unit: 8.5Khu
◆Maximum tube voltage: 75Kv
◆Target angle: 21°
◆Total filtration: 1.6mmAL (Origin filtration: 0.8mmAL, Additional filtration: 1mmAL)
◆X-ray irradiation range: 73mm (roundness)
e) Useage voltage: battery DC16.8V
f) Charger: Input AC100-240V, Output: DC16.8V
Use x-ray unit MSLK04 Product Composition:
1. Fixed tie
2. Power soft-switching
3. Subtraction adjustment
4. Work start button
5. Recharge Stand
6. External start
7. The main power switch
9. The open of battery replacement
10. Status display
11. Battery charge
12. Work time
13. Units level degrees
14. Teeth straight edge
Developing the Dental film
1. Picture the tooth following the right operating procedure of the X-ray Unit，
then take the film out from the mouth.
2. Use a syringe to extract 4-5 mi of imaging liquid.
3. Transfuse the liquid into the bag，nip both the ends and the sides of the bag
with thumbs and first fingers(knead the film).
4. Open the end of the bag about ten minutes after transfusing the liquid into the bag.
5. Take the film out，put it into clear water，and rinse it. Don't wipe the surface of the
film with the surface to prevent damage to the image.
6. Observe and diagnose after the film becomes dry.
Use x-ray unit MSLK01 Use of the Equipment
The machine is made up of two parts: the main unit and the ball head. The main unit includes the
power supply，central data processing unit，and the part to shock，enlarge，and feedback.
The ball head is made up by components with functions of controlling pressure and frequency
and ball tube，vacuum sealed part， etc. The shell is from an injection mold with lead sealing protection.
The machine is mainly used in dental clinics，to check out before treatment，and to be used for
comparison and confirmation during and after treatment. It is an indispensable piece of equipment
for daily work within a clinic. Some examples of its use include checking tissue structure，the depth
of the root，the degree of inflammation，and for the inner inspection of broken teeth.
While using a small amount of radiation there is no need for protection. Hospitals have purchased this
machine to take films of fingers and toes by the side of a patient's bed.
MSL TEAM picture
MSL Medical cooperate with DHL,FEDEX,UPS,EMS,TNT,etc.International shipping company,make your goods arrive destination safely and quickly.