Good Price Vein Finder MSL-263
MSL Good Price Vein Finder MSL-263 Operating Principle
Handheld portable infrared vein finder acquires image of subcutaneous veins,the image which
results from dealing with image signal is projected onto the surface of the skin.Thus, subcutaneous
vein image will be displayed on the skin surface of the corresponding position.
Cheap Good Price Vein Finder MSL-263 Technical Parameter
Effective positive projection distance: 29cm～31cm
Light projection: 300lux～1000lux
Active radiation contains wavelength light: 750nm～980nm
Electrical source: lithium ion polymer batteries
service voltage: d.c. 3.0V～4.2V
Net weight: 0.28±0.02kg
Prevent into the liquid level: IPX0
Best Good Price Vein Finder MSL-263 Method of Application
Before using the product, medical staff should be familiar with how to correctly locate in
order to get the best detection effect. In the process of use, it is recommended that the
medical staff take touch and visual observation information for reference, combined
with the projected image results to determine the actual location of veins. Please try
to keep the observed of skin smooth and tight, give priority to choose areas of the
organization with less scars ,spots or hair.
1. Start The Instrument
a. Press and release the power button of the panel, then the instrument starts.
b. After the instrument starts up,power indicator lights up. According to the power quantity ,
the light shows two colors: white, blue.White represents high power, low blue stands for low power.
c. Please make sure that the power indicator is normal and power is full while the instrument works.
Power indicator displaying blue indicates the instrument needs to be recharged in time. When the
instrument is charging, charging indicator is flashing white light. If it is fully charged, charging indicator puts out.
2. Repeater Operation
a. Distance measurement: Pull the antenna out to maximum, make the antenna perpendicular to
the surface of the skin and the antenna head contact the surface of the skin.You can observe
instrument square image projected onto the surface of the skin.The grain of the image is corresponding
with subcutaneous vein grain. Medical staff choose the appropriate target vein to complete the venipuncture.
b. Imaging mode switch:Insert the needle into the hole of projection window, press imaging mode
switch button ,image can be switched to blue-white,red-green and red-white imaging mode.
c. Image brightness adjustment: Touch image brightness adjustment button to adjust image
brightness.”” means brightness enhancement, "" means brightness reduction.
d. Sleep:The instrument starts sleep mode when touch sleep button.The image is off. Meanwhile ,
the instrument saves electricity. If you touch the sleep button again, the image restores.
3. Shut down
After using, press the power button, the instrument is turned off and power indicator is extinguished.
4. Suggestions for use
a. Clinical use: Two methods for clinical use are recommended.The first method is that medical
staff begin venipuncture directly while using the instrument .The second one is to draw lines towards
along the projection image of targeted veins by a medical skin marker.It is convenient for
venipuncture after shutdown.
b. Sunlight or indicatorlight illume: While you are using the instrument, please do not make
the measured surface of skin towards to the sun. Otherwise the vein image is incomplete.
It is necessary to adjust the indoor illumination which can help medical staff to examine the vein display better.
Attention and Caution
1. Handheld portable infrared vein finder is a kind of medical equipment which it forms the
subcutaneous vein image in situ for contactless.
2. In order to accurately examine the position of the vein, it is suggested put the product
at the proper height and angle, and keep the product located in the center of the target vein.
3. Don’t look straight at the light source when it is working.
4. This product belongs to the electronic equipment. There could be outside interference by
electromagnetic signal.So please stay away from other equipments when using.
5. It is suggest that the instrument should not be used when it is charging.
6. The instrument has no waterproof function, please keep it from liquid.
7. Please do not open, disassemble or repair the instrument by youself.
8. If the product is expected to be not used for a long time, please charge the product fully,
clean and package it in a dry, shady and cool place to store by the original packaging materials.
Please avoid putting the product upside down and under heavy stuff when storaging.
9. This product contains lithium polymer batteries, it is strictly prohibited to put the product into the fire.
Don't discard at will and contact the manufacturer for recycling.
1. It is suggested to maintain regular equipment, ensure it enough clean before use.
2. Attention for instrument maintenance matters:
a. The instrument has no waterproof function, please keep it from water and don’t operate with wet hands.
b. Please do not use the method of ultraviolet irradiation or high temperatures for disinfection.
c. It is suggested that do not charge the instrument when it is maintained.
d. You can disinfect the instrument by a clean dry cloth which be wet with soap, medicinal alcohol and twisted dry.
Store in the cool, dry, dark place where temperature is between 5℃ to 40℃ and relative humidity does not exceed 80%.
MSL factory picture，medical supplier for Long-term cooperation.
MSL TEAM picture
MSL Medical cooperate with DHL,FEDEX,UPS,EMS,TNT,etc.International shipping company,make your goods arrive destination safely and quickly.