10,000 oxygen machine of units delivered India. Best Oxygen Generator Price in China. High oxygen purity. Low noise. Easy to use. Delivers up to 95.5% oxygen concentration. OEM&ODM serive、In stock, Fast Shipping、Free Customs Clearance. Medical grade,Dual flow 10L, 10L oxygen concentrator for sale. Yuwell oxygen machineProvide the best price.
Net weight: about 29kg
Dimensions: 364*385*731 (mm)
The length of the oxygen tube should not exceed 15.2 meters
|Packaging detail:Standard export package
Delivery detail:within 7-10 workdays after receipt of payment
1. Maximum recommended flow rate: 10L/min
2. Flow range when the outlet nominal pressure is 7kPa: 1——10L/min
3. When the maximum recommended flow rate is applied, a back pressure
of 7kPa is applied, and the flow rate changes: <1L/min
4. The oxygen concentration when the outlet nominal pressure is zero (reach the concentration level within 30 minutes of initial startup): Oxygen flow
rate 1——10L/min, oxygen concentration: 93%±3%
5. Output pressure: 30——70kPa
6. Compressor relief valve release pressure: 250kPa±50 kPa
7. Noise of the whole machine: <60dB (A)
8. Power supply: AC220V/50Hz
9. Input power: 550VA
11. Net weight: about 29kg
12. Dimensions: 364*385*731 (mm)
13. Altitude: The oxygen concentration will not be reduced when the sea level reaches 1828 meters, and the efficiency is less than 90% when the sea level reaches from 1828 meters to 4000 meters.
14. Security system:
If the current is overloaded or the connection line is loose, the machine will stop;
Compressor high temperature, shutdown;
Power off, alarm and shutdown;6
15. Minimum working time: no less than 30 minutes;
16. Electrical classification: Class Ⅱ equipment, type B application part;
17. Work system: continuous operation
18. Normal working environment:
Ambient temperature range: 10℃——40℃;
Relative humidity≤80% range;
Atmospheric pressure range: 860hPa-1060 hPa;
Note: When the storage and transportation temperature is lower than 5°C, the equipment should be placed in a normal working environment for more than four hours before use.
19. Oxygen outlet temperature ≤46℃;
20. Recommendation: The length of the oxygen tube should not exceed 15.2 meters, and it should not be collapsed;
21. Liquid-proof grade: IPXO
22. Equipment type: non-AP/APG equipment (cannot be used with flammable anesthetic gas
mixed with air or flammable anesthetic gas mixed with oxygen or imine oxide
Thank you for choosing and using our oxygen generator
Medsinglong Global Group Co.,Ltd.
Safety Precautions Special warning:
▲ In order to prevent power outages or possible failures of the oxygen concentrator, those in urgent need of oxygen and critically ill patients must be equipped with other backup oxygen supply devices (such as oxygen cylinders, oxygen bags, etc.)! This device is suitable for oxygen supplementation and is not considered for life support
or life extension!
▲ The nasal oxygen tube is provided separately by the customer, please use the nasal oxygen tube product with the medical device registration certificate.
The warnings and cautions shown here are for the correct and safe use of the product to prevent users or others from causing harm or damage.
Use electromagnetic environment guidance
This machine is suitable for places including hospitals, homes and other buildings directly connected to the civil low-voltage power supply.
The radio frequency energy used by this machine only acts on the internal operation of the machine. Therefore, its radio frequency emission is very low and it is impossible to affect other electrical equipment nearby.
Due to the dramatic increase in the number of radio frequency transmitters or other electrical noise sources in medical care settings, serious interference caused by too close or too strong transmitter power may cause the machine to stop working.
If this happens, check the site to find the source of interference, and take the following measures to eliminate the interference: (1) Turn off nearby equipment and then turn on the unit; (2) Change the direction or location of the interfering equipment; (3) Increase the distance between the interfering equipment and this equipment.
Measures to reduce the risks of burns, electric shocks, fires or human injuries Avoid using the oxygen concentrator when taking a bath. If you need to use it continuously, please follow the doctor’s prescription, and the oxygen concentrator must be used in another room 2.5 meters from the bathroom.
Do not place or store the oxygen generator in a place where water or other liquids can drip easily.
If the oxygen generator falls into the water, do not touch it, immediately cut off the power supply, and contact a qualified dealer or manufacturer.
Never leave the oxygen generator unattended after the power is turned on.
Please refer to this manual for the use of the oxygen generator. If the user or service personnel feel that the amount of oxygen is insufficient, please contact the supplier or doctor immediately and adjust the flow according to the doctor's instructions.
When using this product, please supervise it when it is close to children or people with disabilities.
Do not use parts, accessories or adapters that are not approved by the manufacturer. The use of humidifiers and operating accessories not specified for this oxygen generator will reduce the performance of the machine.
Do not connect this product with other oxygen generators or oxygen therapy equipment in parallel or series.
Oxygen therapy can be dangerous in some specific environments. The manufacturer recommends that users consult a doctor for advice before using this product.
Avoid any sparks near medical oxygen equipment, including sparks caused by various frictional static electricity.
If the power cord or plug of the oxygen generator is damaged, if the machine does not work properly, if the machine is dropped and damaged, please contact qualified maintenance personnel for inspection and repair.
If an accident occurs during the use of the product, call the emergency hotline immediately and seek the help of professional medical personnel.
The safe use period of this product is 26280 hours from the date of use. Keep the power cord away from heated or hot surfaces.
Do not move the oxygen generator when it is charged.
Do not drip or insert any substance into the opening of the machine.
Product name: Small medical oxygen generator Product model specification: MSLBB205
Factory number, production date: see product label Product technical indicators: